1. Spectroscopic setup (15 points)

Arriving at the
observatory for my spectroscopic observing run, I find that the only
grating I have available is a 300 line/mm grating blazed at 8000A in
the first order. But I want to observe in the blue from 3500 to
5000A.

- How could I best do this and what precautions should I take?
- At what angle will 4000A light be found relative to the grating normal?
- What is the blaze angle of the grating?
- What camera focal length should I use to fit the spectrum on a CCD 20mm wide?
- If the 2 arcsecond wide slit projects to 66 microns on the detector, what is the spectral resolution (give it in terms of both delta-lambda and R)?
- What is the velocity resolution (in km/s)?

For this problem, assume the grating surface is
normal to light from the collimator.

2. Slit positioning (10 points for 306, 20 points for 406)

Now, using the parameters from problem #1, think about the effect of moving a star from one edge of the slit to the other.

- If stars were truly point sources (unblurred by atmospheric seeing or telescope optics), what would be the velocity change in the star that you would observe if you moved the star from one edge of the slit to the other?
- Conceptually, how would you expect the velocity to change from one side of the slit to the other if the seeing was extremely good, say 0.5 arcsec FWHM? What if it was awful, say 5.0 arcsec FWHM? How would it change as the seeing varied in between those extremes? Explain your reasoning; a sketch might help your explanation.

ASTR 406: Make a plot
of the velocity difference as a function of FWHM for seeing
that ranges from FWHM=0.1 arcsec (unrealistically good!) to 5 arcsec
(utterly horrible). Make the simplifying assumption that in the spatial
direction we are only extracting the spectrum centered on the star, so
that you can treat the star's profile in the slit as a one dimensional
gaussian. Hint: you'll probably need to make use of the
scipy.special.erf() function at some point in making this plot!

3. Wavelength choices (5 points)

If you have a spectrograph (not the one in the previous problems) where you can observe at
either 4000A or 8000A with the same resoution (FWHM = 2
Angstroms), which of these two wavelength settings will give the
better velocity accuracy all other things being equal?

4. Spectrograph design (15 points)

You designing a spectrograph for a 5m telescope at f/10 Cass focus. The collimator is a 6 inch collimator and the slit is 1 arcsec wide and projects to 30 microns (3pixels) on the detector. An astronomer askes you what spectral resolution will she will have with a 1200 line/mm grating at 5000A. What do you tell her? Explain any assumptions you need to make.

No more problems will be coming......